The sample is installed onto PTFE sample support outside the glass cell and is connected with electrolyte through the area of 1cm2 hole. Sample mounting screws provided on the frame ensure the rigid mounting of the sample without electrolyte leakage through the hole. The PTFE frame of the other side of the glass-cell contains platinum-mesh as a counter electrode (included parts). Kanopy reference electrodes (not included, can be ordered separately) will fit in the B-14 neck joint adjacent to the working electrode. This setup comes with extra neck joints the can be used either for gas-purging or temperature sensor or simply kept closed with a glass stopper when unused.
|Working Volume||A borosilicate glass cell of 250mL volume; customized volume is available.|
|Working electrode port||1cm2 bore onto the PTFE frame, which exposes the working electrode/sample to the electrolyte solution.|
|Working electrode dimensions||Minimum 12mm X 12mm.|
|Sample mounting screw||A threaded rod to attach the coupon tightly on the working electrode port.|
|Reference electrode port||SGJ B14, compatible with all Kanopy reference electrodes.|
|Counter electrode port||A small bore onto the PTFE frame containing Pt-mesh counter electrode of mesh size 10mmx10mm.|
|Spare port||SGJ B14; can be used for other purposes such as gas purging or temperature sensing.|